Payment & Shipping Terms:
|Application:||Vickers Hardness Tester||Model:||IVicky 2.0|
|Camera:||1.3MP USB Camera||OS:||Win 7, 8, 10|
|Report:||In Word, Excel Format||Image:||Can Be Saved In Report|
Automatic Knoop Vickers Hardness Measurement Software Generate Report With Usb Camera
SINOWON new generation Vickers (Micro Vickers)/Knoop measurement software iVicky with friendly
operation interface, apply USB camera to fast capture video and images, supports case depth inspection by analyzing the indentation image.
In traditional way, operators judge the size of the indentation through the microscopes, due to vision errors, different operators obtain different measurement results for same indentation. While by software, the indentation is enlarged through high resolution lens, and the size of indentation can be automatically obtained by image processing algorithm.
Following are highlighting of iVicky:
1, Easy Calibration, calibration can be done according to standard hardness blocks. Also can select calibration data according to test force, this function need to keep same test force with hardness tester.
2, Auto Measurement: Apply Digital image processing technology, software can automatically find indentation precisely and compute hardness value. Manually adjust indentation length also available if indentation is not clearly.
3, Manual Measurement: Support two methods for manual measurement, click four peaks of indentation or measure length of diagonal line.
4, support hardness conversion among 12 hardness scales according to ASTM E140-05, hardness results correction of cylinder surface, upper and lower limit setting.
5, Hardness Results Display: Shows indentation dialog length, conversion value, requested minimal sample thickness, minimal any two indentations distance and material edge to the center of indentation.
6, Test Report, export test results with indentation to MS EXCEL or WORD
7, Statistics: Automatically compute average value, variance, etc.
8,Others Functions: Support manual measurement, delete measurement result, etc.
Contact Person: Mr. Marshall Zheng