Payment & Shipping Terms:
|Hardness Scales :||Vickers(HV);Rockwell C(HRC); Rockwell A(HRA); RockwellB(HRB);Brinell(HB)||Hardness Conversion :||HV HK, HRA, Etc|
|Data Output :||RS232 Hyper Terminal/Bluetooth Printer||Data Display:||LCD Backlight, Adjustable|
|Data Storage:||To Save 1000-groups Of Measuring Data And 20-groups Of Calibration Data||Language:||Chinese, English, German, Portugese, Turkish|
|Battery Voltage:||Rechargeable Lithium Battery ; Voltage :4.2V, 4800mAh||Recharger Voltage:||AC220V/50Hz, 110V/60Hz|
Portable Hardness Testing Equipment,
Portable Hardness Testing Machine
60Hz Portable Hardness Tester metal hardness testing equipment Universal impact direction
Ultrasonic Hardness Testing apply UCI method: Ultrasonic Contact Impedance, a hardness testing method developed by Dr. Claus Kleesattel in 1961 based on the measurement of the frequency shift of a resonating rod caused by the essentially elastic nature of the finite area of contact between the indenter and the test piece during the penetration, the execute standard is ASTM A1038.
Optional Manual Probes
Optional Testing Stand
When selecting a hardness tester for your application, it is important for you to consider the following:
A, Choose the correct test method based on the application.
Plan to use the highest test force and largest indenter possible. Consider the effects of the shape and dimensions of your test sample.
B, Verify the test results meet your requirements for accuracy and repeatability. There are significant differences between levels of performance within each classification of tester. A difficult job on one tester could be very simple and fast on another.
Answer these key questions:
1, What kind of hardness scale to be used?
2, What is the material to be tested, and is this material suitable to the type of test method you are considering?
3, How large is the part, component or specimen to be tested?
4, Is the test point difficult to reach?
5, What is the quantity of testing that will be done?
6, How accurate does your test result need to be?
7, What is your budget?
8, What is the required return on investment and do you have ways to measure reductions in costs- yields, throughput, operator efficiency?
9, What testing problems have you experienced in your current method?
10, How knowledgeable are the users of the tester?
Technical Specifications of SU-300M:
|Product Name||Ultrasonic Hardness Tester|
|Optional probe||Manual Probe:10N,20N,49N,98N ;
|Measuring Range (ASTM)||HBS:100-500; HV:100-1500; HRC:20-68; HRB:55-100;
HRA:37-85; Mpa: 255-2180N/mm
|Hardness Scales||Vickers(HV);Rockwell C(HRC); Rockwell A(HRA);
|Data Display||LCD Backlight, Adjustable|
|Language||Chinese, English, German, Portugese, Turkish|
|Data Output||RS232 Hyper Terminal/Bluetooth Printer|
|Data Storage||To save 1000-groups of measuring data and 20-groups of calibration data|
|Word input||Letters and numerals|
|Working Environment||Temperature: -10°C to 50°C ; Humidity :30%-80% R.H.|
|Battery Voltage||Rechargeable lithium battery ; Voltage :4.2V, 4800mAh|
|Recharger Voltage||AC220V/50Hz, 110V/60Hz|
|Main unit Dimension and Weight||Main unit Dimension :160 x 80 x 30mm (L x W x H);Manual
Probe:Φ22×153mm ; Weight(No Probe): 0.5Kg
|Packing Dimension and Weight||Packing Dimension 320 x 430 x 155mm (L x W x H) ;
Packing Weight (Standard Delivery)4.3Kg
|Product Name||Code #||Product Name||Code #||Product Name||Code #|
|Instrument Main Body||882-141M||10N Motorized Probe||882-251||Anti Vibration Case||882-901|
|Standard Rockwell Block||882-641||Probe Cable||882-801||Screw||882-951|
|USB Recharger||882-851||COM Cable||882-810||Operation Manual||SU-300M|
|Qualification Certificate||SU-300M||Packing List||SU-300M||Warranty Card||SU-300M|
Contact Person: Mr. Marshall Zheng